<?xml version="1.0"?>
<GCContextualMethod xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance" xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/">
  <method MethodName="" xmlns="">
    <GC PostRunHoldTime="0" AutoPrepRun="UNKNOWN">
      <Inlet DeterminesReadiness="true">
        <Setpoints COCTrackOven="false" SeptumPurgeFlow="3" SeptumPurgeState="ON" SeptumPurgeSwitching="false" SeptumPurgeSwitchTime="0" TotalFlow="38.699999999220815" TotalFlowState="ON" ControlMode="PRESSURE">
          <Temperature State="ON" InitialHoldTime="0" InitialValue="250" PostRunValue="250" />
          <Pressure State="ON" InitialHoldTime="0" InitialValue="306488.61589999986" PostRunValue="758423.27" />
          <Cryo UseCryo="UNKNOWN" UseTemp="0" TimeoutDetection="UNKNOWN" Timeout="0" FaultDetection="UNKNOWN" QuickCool="UNKNOWN" />
          <GasSaver Time="2" Flow="20" State="ON" />
          <Operation Mode="SPLIT" SplitRatio="50" PurgeFlow="40" PurgeTime="0.5" PulsePressure="689475" PulseTime="0.75" VIInjectTime="0" VentFlow="0" VentPressure="0" VentTime="0" />
          <Temperature2 State="ON" InitialHoldTime="0" InitialValue="250" PostRunValue="250" />
          <PreRunFlowTest PreRunFlowTestOnOff="OFF" PreRunFlowTestFailureAction="CONTINUE" />
        </Setpoints>
      </Inlet>
      <Inlet DeterminesReadiness="false" />
      <Column DeterminesReadiness="true">
        <Setpoints Mode="CONSTANT_FLOW">
          <Flow State="ON" InitialHoldTime="0" InitialValue="0.699999999984722" PostRunValue="1" />
        </Setpoints>
      </Column>
      <Column DeterminesReadiness="true">
        <Setpoints Mode="CONSTANT_PRESSURE">
          <Flow State="ON" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
        </Setpoints>
      </Column>
      <Column DeterminesReadiness="true">
        <Setpoints Mode="CONSTANT_PRESSURE">
          <Flow State="ON" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
        </Setpoints>
      </Column>
      <Column DeterminesReadiness="true">
        <Setpoints Mode="CONSTANT_PRESSURE">
          <Flow State="ON" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
        </Setpoints>
      </Column>
      <Column DeterminesReadiness="true">
        <Setpoints Mode="CONSTANT_PRESSURE">
          <Flow State="ON" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
        </Setpoints>
      </Column>
      <Column DeterminesReadiness="true">
        <Setpoints Mode="CONSTANT_PRESSURE">
          <Flow State="ON" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
        </Setpoints>
      </Column>
      <Oven DeterminesReadiness="true" OverideColumnMaxTemp="false">
        <Setpoints UseCryo="OFF" UseTemp="25" TimeoutDetection="OFF" Timeout="120" FaultDetection="OFF" QuickCool="OFF" ColdTrapTemp="0" EquilibrationTime="1">
          <Temperature State="ON" InitialHoldTime="3.2" InitialValue="40" PostRunValue="50">
            <Ramp Rate="25" Next="250" HoldTime="13.4" />
          </Temperature>
        </Setpoints>
      </Oven>
      <Detector DeterminesReadiness="true">
        <Setpoints Temperature="300" Heater="ON" FuelFlow="30" Fuel="ON" UtilityFlow="400" Utility="ON" MakeupFlow="20" Makeup="ON" ConstantTotalFlow="false" TotalFlow="0" ConstantTotalFuel="true" Electronics="ON" Flame="ON" State="UNKNOWN" ColComp1="0" ColComp2="0">
          <Performance>
            <BlankRunAnalysis>
              <PerformAnalysis>false</PerformAnalysis>
              <InitialBaseline MaxValue="20" MinValue="2" NoiseValue="0.3" FailureAction="CONTINUE" />
              <FinalBaseline MaxValue="40" MinValue="2" NoiseValue="0.6" FailureAction="CONTINUE" />
              <PeakAnalysis MaxTotalPeakArea="100" MaxPeakHeight="3" FailureAction="CONTINUE">
                <TimeWindow Start="0" End="8.2" />
              </PeakAnalysis>
            </BlankRunAnalysis>
          </Performance>
        </Setpoints>
      </Detector>
      <Detector DeterminesReadiness="false" />
      <Detector DeterminesReadiness="false" />
      <Detector DeterminesReadiness="false" />
      <AuxThermalZone DeterminesReadiness="false" />
      <AuxThermalZone DeterminesReadiness="false" />
      <AuxThermalZone DeterminesReadiness="false" />
      <PCM>
        <ChannelA DeterminesReadiness="false" />
        <ChannelB DeterminesReadiness="false" />
      </PCM>
      <PCM>
        <ChannelA DeterminesReadiness="false" />
        <ChannelB DeterminesReadiness="false" />
      </PCM>
      <PCM>
        <ChannelA DeterminesReadiness="false" />
        <ChannelB DeterminesReadiness="false" />
      </PCM>
      <PCM>
        <ChannelA DeterminesReadiness="false" />
        <ChannelB DeterminesReadiness="false" />
      </PCM>
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <Valve InitialState="UNKNOWN" />
      <CryoFocus PreRunTime="0" HoldTime="0" />
      <Signals P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="UNKNOWN" P3="5" index="0" />
      <MSD>
        <TransferLine State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
        <Bus State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
      </MSD>
      <OverlapSettings OverlapOnOff="OFF" OverlapTime="0" OverlapMode="AFTER_PREV_INJECTION" />
      <ChromatographicAttributes Enabled="false">
        <Signal Enabled="true" SignalIndex="0">
          <Parameters CheckoutSample="FID" />
        </Signal>
      </ChromatographicAttributes>
    </GC>
    <Sampler>
      <FrontInjector Fan="ON" injmode="NORMAL">
        <PreWashA Reps="3" Volume="80" DrawSpeed="200" DispenseSpeed="20" ViscosityDelay="2" />
        <PreWashB Reps="3" Volume="80" DrawSpeed="200" DispenseSpeed="20" ViscosityDelay="2" />
        <SampleWash Reps="2" Volume="20" DrawSpeed="200" DispenseSpeed="20" ViscosityDelay="2" DepthOffset="0" />
        <SamplePumps Reps="6" DrawSpeed="200" DispenseSpeed="20" ViscosityDelay="2" DepthOffset="0" />
        <Sample Volume="10" DrawSpeed="200" ViscosityDelay="2" DepthOffset="0" Airgap="2" />
        <Sample Volume="10" DrawSpeed="200" ViscosityDelay="2" DepthOffset="0" Airgap="2" />
        <Sample Volume="10" DrawSpeed="200" ViscosityDelay="2" DepthOffset="0" Airgap="2" />
        <Injection DispenseSpeed="10" PreInjectionDwellTime="0" PostInjectionDwellTime="0" LargeVolumeReps="1" LargeVolumeDelay="0" />
        <PostWashA Reps="3" Volume="80" DrawSpeed="200" DispenseSpeed="20" ViscosityDelay="2" />
        <PostWashB Reps="3" Volume="80" DrawSpeed="200" DispenseSpeed="20" ViscosityDelay="2" />
      </FrontInjector>
      <BackInjector Fan="OFF" injmode="NORMAL">
        <PreWashA Reps="0" Volume="20" DrawSpeed="200" DispenseSpeed="10" ViscosityDelay="0" />
        <PreWashB Reps="0" Volume="20" DrawSpeed="200" DispenseSpeed="10" ViscosityDelay="0" />
        <SampleWash Reps="0" Volume="20" DrawSpeed="200" DispenseSpeed="10" ViscosityDelay="0" DepthOffset="0" />
        <SamplePumps Reps="6" DrawSpeed="200" DispenseSpeed="10" ViscosityDelay="0" DepthOffset="0" />
        <Sample Volume="10" DrawSpeed="200" ViscosityDelay="0" DepthOffset="0" Airgap="2" />
        <Sample Volume="10" DrawSpeed="200" ViscosityDelay="0" DepthOffset="0" Airgap="2" />
        <Sample Volume="10" DrawSpeed="200" ViscosityDelay="0" DepthOffset="0" Airgap="2" />
        <Injection DispenseSpeed="10" PreInjectionDwellTime="0" PostInjectionDwellTime="0" LargeVolumeReps="1" LargeVolumeDelay="0" />
        <PostWashA Reps="0" Volume="20" DrawSpeed="200" DispenseSpeed="10" ViscosityDelay="0" />
        <PostWashB Reps="0" Volume="20" DrawSpeed="200" DispenseSpeed="10" ViscosityDelay="0" />
      </BackInjector>
      <Tray UseBarcodeReader="false">
        <Mixing Enable="false" Speed="1000" Cycles="2" Intensity="UNKNOWN" Time="2" />
        <BCRHeating Duration="1" Setpt="50" Enable="false" />
      </Tray>
    </Sampler>
  </method>
  <supplement xmlns="">
    <signal P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="UNKNOWN" P3="5" index="0" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="1" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="2" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="3" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="4" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="5" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="6" />
    <signal P0="UNKNOWN" P1="UNKNOWN" P2="UNKNOWN" P3="0" index="7" />
    <signalIsSaved>true</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalIsSaved>false</signalIsSaved>
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalZero Zero="false" />
    <signalTimeEvents>
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
      <SignalEventList xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/" />
    </signalTimeEvents>
    <firstSamplePreferences>1</firstSamplePreferences>
    <AlsErrorAction>PAUSE</AlsErrorAction>
    <enableAlsTray>false</enableAlsTray>
    <methodResolutionCompleted>false</methodResolutionCompleted>
    <overlappedInjectionMode>noOverlap</overlappedInjectionMode>
    <overlappedInjectionDelay>0</overlappedInjectionDelay>
    <InjectorID />
    <useSampleSelectorAsValve>false</useSampleSelectorAsValve>
    <IqMethodSettings>
      <Keys>idx_signal1_szInlDef</Keys>
      <Keys>idx_signal1_fRate</Keys>
      <Values>A0</Values>
      <Values>5</Values>
    </IqMethodSettings>
  </supplement>
  <configuration SleepMethodChanged="false" WakeMethodChanged="false" ConditioningMethodChanged="false" MSVentMethodChanged="false" MSShutdownMethodChanged="false" KeyboardLocked="false" KeyboardLockedExtended="NeverLock" NoPrepOnManual="true" SuppressMethodResolution="false" xmlns="">
    <Headspace xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/">
      <Front Address="" Type="NONE" SerialNumber="" Status="OFF" ConnectTime="0" Present="false" DCOMMEnabled="false" LoopVolume="0" BarcodeReaderPresent="false" CoolingPlatePresent="false" TransferLineType="0" CarrierControlPresent="false" TransferLineDiameter="0" FWRev="" ManufacturingDate="0" xmlns="" />
      <Back Address="" Type="NONE" SerialNumber="" Status="OFF" ConnectTime="0" Present="false" DCOMMEnabled="false" LoopVolume="0" BarcodeReaderPresent="false" CoolingPlatePresent="false" TransferLineType="0" CarrierControlPresent="false" TransferLineDiameter="0" FWRev="" ManufacturingDate="0" xmlns="" />
    </Headspace>
    <Metadata>
      <Maximum>
        <GC PostRunHoldTime="0" AutoPrepRun="UNKNOWN">
          <Inlet DeterminesReadiness="true">
            <Setpoints COCTrackOven="false" SeptumPurgeFlow="30" SeptumPurgeState="UNKNOWN" SeptumPurgeSwitching="false" SeptumPurgeSwitchTime="0" TotalFlow="1250" TotalFlowState="UNKNOWN" ControlMode="FLOW">
              <Temperature State="UNKNOWN" InitialHoldTime="999.99" InitialValue="400" PostRunValue="400">
                <Ramp Rate="0" Next="400" HoldTime="999.99" />
              </Temperature>
              <Pressure State="UNKNOWN" InitialHoldTime="999.99" InitialValue="6894757.29316836" PostRunValue="6894757.29316836">
                <Ramp Rate="10342135.94" Next="6894757.29316836" HoldTime="999.99" />
              </Pressure>
              <Cryo UseCryo="UNKNOWN" UseTemp="0" TimeoutDetection="UNKNOWN" Timeout="0" FaultDetection="UNKNOWN" QuickCool="UNKNOWN" />
              <GasSaver Time="999.99" Flow="1250" State="UNKNOWN" />
              <Operation Mode="SPLITLESS" SplitRatio="12500" PurgeFlow="1250" PurgeTime="999.99" PulsePressure="6894757.29316836" PulseTime="999.99" VIInjectTime="0" VentFlow="1250" VentPressure="0" VentTime="0" />
              <Temperature2 State="UNKNOWN" InitialHoldTime="999.99" InitialValue="400" PostRunValue="400">
                <Ramp Rate="0" Next="400" HoldTime="999.99" />
              </Temperature2>
            </Setpoints>
          </Inlet>
          <Inlet DeterminesReadiness="false" />
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="999.99" InitialValue="1250" PostRunValue="1250">
                <Ramp Rate="100" Next="1250" HoldTime="999.99" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="999.99" InitialValue="1250" PostRunValue="1250">
                <Ramp Rate="100" Next="1250" HoldTime="999.99" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="999.99" InitialValue="1250" PostRunValue="1250">
                <Ramp Rate="100" Next="1250" HoldTime="999.99" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="999.99" InitialValue="1250" PostRunValue="1250">
                <Ramp Rate="100" Next="1250" HoldTime="999.99" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="999.99" InitialValue="1250" PostRunValue="1250">
                <Ramp Rate="100" Next="1250" HoldTime="999.99" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="999.99" InitialValue="1250" PostRunValue="1250">
                <Ramp Rate="100" Next="1250" HoldTime="999.99" />
              </Flow>
            </Setpoints>
          </Column>
          <Oven DeterminesReadiness="false">
            <Setpoints UseCryo="UNKNOWN" UseTemp="0" TimeoutDetection="UNKNOWN" Timeout="0" FaultDetection="UNKNOWN" QuickCool="UNKNOWN" EquilibrationTime="999.99">
              <Temperature State="UNKNOWN" InitialHoldTime="999.99" InitialValue="450" PostRunValue="450">
                <Ramp Rate="120" Next="450" HoldTime="999.99" />
              </Temperature>
            </Setpoints>
          </Oven>
          <Detector DeterminesReadiness="false">
            <Setpoints Temperature="450" Heater="UNKNOWN" FuelFlow="100" Fuel="UNKNOWN" UtilityFlow="800" Utility="UNKNOWN" MakeupFlow="100" Makeup="UNKNOWN" ConstantTotalFlow="false" TotalFlow="0" Electronics="UNKNOWN" Flame="UNKNOWN" State="UNKNOWN" />
          </Detector>
          <Detector DeterminesReadiness="false" />
          <Detector DeterminesReadiness="false" />
          <Detector DeterminesReadiness="false" />
          <AuxThermalZone DeterminesReadiness="false" />
          <AuxThermalZone DeterminesReadiness="false" />
          <AuxThermalZone DeterminesReadiness="false" />
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <GCxGCValve SampleTime="0" ModulationTime="0" ValveIdleState="UNKNOWN" DelayTime="0" />
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="999.99" InjectTime="999.99" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN" />
          <Valve InitialState="UNKNOWN" />
          <CryoFocus PreRunTime="0" HoldTime="0" />
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
        </GC>
      </Maximum>
      <Minimum>
        <GC PostRunHoldTime="0" AutoPrepRun="UNKNOWN">
          <Inlet DeterminesReadiness="true">
            <Setpoints COCTrackOven="false" SeptumPurgeFlow="0" SeptumPurgeState="UNKNOWN" SeptumPurgeSwitching="false" SeptumPurgeSwitchTime="0" TotalFlow="0" TotalFlowState="UNKNOWN" ControlMode="FLOW">
              <Temperature State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0">
                <Ramp Rate="0" Next="0" HoldTime="0" />
              </Temperature>
              <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0">
                <Ramp Rate="0" Next="0" HoldTime="0" />
              </Pressure>
              <Cryo UseCryo="UNKNOWN" UseTemp="0" TimeoutDetection="UNKNOWN" Timeout="0" FaultDetection="UNKNOWN" QuickCool="UNKNOWN" />
              <GasSaver Time="0" Flow="15" State="UNKNOWN" />
              <Operation Mode="SPLITLESS" SplitRatio="0.1" PurgeFlow="0" PurgeTime="0" PulsePressure="0" PulseTime="0" VIInjectTime="0" VentFlow="0" VentPressure="0" VentTime="0" />
              <Temperature2 State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0">
                <Ramp Rate="0" Next="0" HoldTime="0" />
              </Temperature2>
            </Setpoints>
          </Inlet>
          <Inlet DeterminesReadiness="false" />
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="0" InitialValue="-1250" PostRunValue="-1250">
                <Ramp Rate="0" Next="-1250" HoldTime="0" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="0" InitialValue="-1250" PostRunValue="-1250">
                <Ramp Rate="0" Next="-1250" HoldTime="0" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="0" InitialValue="-1250" PostRunValue="-1250">
                <Ramp Rate="0" Next="-1250" HoldTime="0" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="0" InitialValue="-1250" PostRunValue="-1250">
                <Ramp Rate="0" Next="-1250" HoldTime="0" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="0" InitialValue="-1250" PostRunValue="-1250">
                <Ramp Rate="0" Next="-1250" HoldTime="0" />
              </Flow>
            </Setpoints>
          </Column>
          <Column DeterminesReadiness="false">
            <Setpoints Mode="CONSTANT_FLOW">
              <Flow State="UNKNOWN" InitialHoldTime="0" InitialValue="-1250" PostRunValue="-1250">
                <Ramp Rate="0" Next="-1250" HoldTime="0" />
              </Flow>
            </Setpoints>
          </Column>
          <Oven DeterminesReadiness="false">
            <Setpoints UseCryo="UNKNOWN" UseTemp="0" TimeoutDetection="UNKNOWN" Timeout="0" FaultDetection="UNKNOWN" QuickCool="UNKNOWN" EquilibrationTime="0">
              <Temperature State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0">
                <Ramp Rate="0" Next="0" HoldTime="0" />
              </Temperature>
            </Setpoints>
          </Oven>
          <Detector DeterminesReadiness="false">
            <Setpoints Temperature="0" Heater="UNKNOWN" FuelFlow="0" Fuel="UNKNOWN" UtilityFlow="0" Utility="UNKNOWN" MakeupFlow="0" Makeup="UNKNOWN" ConstantTotalFlow="false" TotalFlow="0" Electronics="UNKNOWN" Flame="UNKNOWN" State="UNKNOWN" />
          </Detector>
          <Detector DeterminesReadiness="false" />
          <Detector DeterminesReadiness="false" />
          <Detector DeterminesReadiness="false" />
          <AuxThermalZone DeterminesReadiness="false" />
          <AuxThermalZone DeterminesReadiness="false" />
          <AuxThermalZone DeterminesReadiness="false" />
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <PCM>
            <ChannelA DeterminesReadiness="false" />
            <ChannelB DeterminesReadiness="false" />
          </PCM>
          <GCxGCValve SampleTime="0" ModulationTime="0" ValveIdleState="UNKNOWN" DelayTime="0" />
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN">
            <GasSamplingValve LoadTime="0.01" InjectTime="0.01" />
            <MultipositionValve Position="0" />
          </Valve>
          <Valve InitialState="UNKNOWN" />
          <Valve InitialState="UNKNOWN" />
          <CryoFocus PreRunTime="0" HoldTime="0" />
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
          <PSDEPC PurgeFlow="0" Purge="UNKNOWN">
            <Pressure State="UNKNOWN" InitialHoldTime="0" InitialValue="0" PostRunValue="0" />
          </PSDEPC>
        </GC>
      </Minimum>
      <Default>
        <GC PostRunHoldTime="0" AutoPrepRun="UNKNOWN">
          <Oven DeterminesReadiness="false" />
          <Detector DeterminesReadiness="false">
            <Setpoints Temperature="0" Heater="UNKNOWN" FuelFlow="0" Fuel="UNKNOWN" UtilityFlow="0" Utility="UNKNOWN" MakeupFlow="0" Makeup="UNKNOWN" ConstantTotalFlow="false" TotalFlow="0" Electronics="UNKNOWN" Flame="UNKNOWN" State="UNKNOWN">
              <Performance>
                <BlankRunAnalysis>
                  <PerformAnalysis>true</PerformAnalysis>
                  <InitialBaseline MaxValue="20" MinValue="2" NoiseValue="0.3" FailureAction="CONTINUE" />
                  <FinalBaseline MaxValue="40" MinValue="2" NoiseValue="0.6" FailureAction="CONTINUE" />
                  <PeakAnalysis MaxTotalPeakArea="100" MaxPeakHeight="3" FailureAction="CONTINUE" />
                </BlankRunAnalysis>
              </Performance>
            </Setpoints>
          </Detector>
        </GC>
      </Default>
    </Metadata>
    <ConnectionSettings>
      <TypeOfConnection>LAN</TypeOfConnection>
      <ConnectionSpecification>192.168.254.11</ConnectionSpecification>
      <ProxyUrl />
      <GCName />
      <GCNotes />
      <SerialNumber>US1922A009</SerialNumber>
      <ModelName>Agilent 8890</ModelName>
      <OwnershipTimeout>-1</OwnershipTimeout>
      <WsdlVersion>2022-03-21</WsdlVersion>
    </ConnectionSettings>
    <GCConfiguration>
      <Fixed CryoType="NONE" BaxterConfigured="false" HydrogenSensorPresent="false">
        <FWRevision BootLoader="BootLoader" Manufacturing="Manuf" GCApplication="2.5.1.9" FPGA="FPGA" Font="Font" />
        <ALS>
          <Controller Model="ALSController" FirmwareRev="ALSControllerFW" ManufacturingID="[NoMfgID]" Present="true" />
          <FrontInjector Model="G4513A" FirmwareRev="A.11.06" ManufacturingID="CN19070165" Present="true" Mounted="true" Aligned="true" DoorClosed="true" CompatibleWithTray="false" TurretMaxSamples="16" NumberOfSampleStops="FIFTY" SyringeVolume="10" />
          <BackInjector Model="" FirmwareRev="" ManufacturingID="" Present="false" Mounted="false" Aligned="false" DoorClosed="false" CompatibleWithTray="false" TurretMaxSamples="0" NumberOfSampleStops="UNKNOWN" SyringeVolume="0" />
          <Tray Model="" FirmwareRev="" ManufacturingID="" Present="false" Positions="0" BCRPresent="false" />
        </ALS>
        <Inlet ManufacturingID="Q119B626" ManufacturingDate="0" Address="1" ConfiguredForGC="true" FWRevCode="1282" EEPROMRevCode="768" TypeID="SS_100" CryoPresent="false" Capped="false">
          <FlowLimitsTable>
            <Table Channel="2" Gas="N2" MaxFlow="500" MinFlow="0" />
            <Table Channel="2" Gas="H2" MaxFlow="1250" MinFlow="0" />
            <Table Channel="2" Gas="He" MaxFlow="1250" MinFlow="0" />
            <Table Channel="2" Gas="ArMe" MaxFlow="200" MinFlow="0" />
            <Table Channel="3" Gas="N2" MaxFlow="30" MinFlow="0" />
            <Table Channel="3" Gas="H2" MaxFlow="30" MinFlow="0" />
            <Table Channel="3" Gas="He" MaxFlow="30" MinFlow="0" />
            <Table Channel="3" Gas="ArMe" MaxFlow="30" MinFlow="0" />
          </FlowLimitsTable>
          <Heater ConfiguredForGC="true" Power="72.5792840957074" Position="0" />
        </Inlet>
        <Inlet ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" CryoPresent="false">
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </Inlet>
        <Detector>
          <Signal ManufacturingID="G4591-6182019040281" ManufacturingDate="0" Address="8" ConfiguredForGC="true" FWRevCode="1282" EEPROMRevCode="768" TypeID="FID" />
          <Signal ManufacturingID="" ManufacturingDate="0" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <EPC ManufacturingID="2119B608" ManufacturingDate="0" Address="6" ConfiguredForGC="true" FWRevCode="1282" EEPROMRevCode="768" TypeID="FID_EPC">
            <FlowLimitsTable>
              <Table Channel="1" Gas="H2" MaxFlow="100" MinFlow="0" />
              <Table Channel="2" Gas="Air" MaxFlow="800" MinFlow="0" />
              <Table Channel="3" Gas="N2" MaxFlow="100" MinFlow="0" />
              <Table Channel="3" Gas="He" MaxFlow="100" MinFlow="0" />
            </FlowLimitsTable>
          </EPC>
          <Heater ConfiguredForGC="true" Power="72.8808865071303" Position="2" CryoValve="0" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </Detector>
        <Detector>
          <Signal ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Signal ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </Detector>
        <Detector>
          <Signal ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Signal ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </Detector>
        <Detector>
          <Signal ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Signal ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </Detector>
        <APC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <APC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <APC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <PCM>
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </PCM>
        <PCM>
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </PCM>
        <PCM>
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </PCM>
        <PCM>
          <EPC ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </PCM>
        <PSDEPC ManufacturingID="" ManufacturingDate="0" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <PSDEPC ManufacturingID="" ManufacturingDate="0" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <PSDEPC ManufacturingID="" ManufacturingDate="0" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <PSDEPC ManufacturingID="" ManufacturingDate="0" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <Valve ConfiguredForGC="false" />
        <AuxHeater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        <AuxHeater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        <AuxHeater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        <GCxGCValve ManufacturingID="" ManufacturingDate="946771200" Address="0" ConfiguredForGC="false" FWRevCode="0" EEPROMRevCode="0" TypeID="UNKNOWN" />
        <ValveBox ConfiguredForGC="false">
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
          <Heater ConfiguredForGC="false" Power="0" Position="0" CryoValve="0" />
        </ValveBox>
        <CryoFocus ConfiguredForGC="false" Valve="0" />
        <AvailableSignals P0="TEST_PLOT" P1="DATA_CHANNEL" P2="UNKNOWN" P3="0" />
        <AvailableSignals P0="ATMOSPHERIC_PRESSURE" P1="UNKNOWN" P2="UNKNOWN" P3="0" />
        <AvailableSignals P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="UNKNOWN" P3="0" />
        <AvailableSignals P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="COMPENSATION_PROFILE1" P3="0" />
        <AvailableSignals P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="COMPENSATION_PROFILE2" P3="0" />
        <AvailableSignals P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="COMPENSATION_PROFILE3" P3="0" />
        <AvailableSignals P0="FRONT_SIGNAL" P1="DATA_CHANNEL" P2="COMPENSATION_PROFILE4" P3="0" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="TEMPERATURE" P2="SETPOINT" P3="0" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="TEMPERATURE" P2="ACTUAL" P3="0" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="TEMPERATURE" P2="POWER" P3="0" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="TEMPERATURE" P2="DUTY_CYCLE" P3="0" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="FLOW" P2="SETPOINT" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="FLOW" P2="ACTUAL" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ACTUAL" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="SETPOINT" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="DUTY_CYCLE" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ADC_ACTUAL" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ADC_SETPOINT" P3="1" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="FLOW" P2="SETPOINT" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="FLOW" P2="ACTUAL" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ACTUAL" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="SETPOINT" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="DUTY_CYCLE" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ADC_ACTUAL" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ADC_SETPOINT" P3="2" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="FLOW" P2="SETPOINT" P3="3" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="FLOW" P2="ACTUAL" P3="3" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ACTUAL" P3="3" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="SETPOINT" P3="3" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="DUTY_CYCLE" P3="3" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ADC_ACTUAL" P3="3" />
        <AvailableSignals P0="FRONT_DETECTOR" P1="PRESSURE" P2="ADC_SETPOINT" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="TEMPERATURE" P2="SETPOINT" P3="0" />
        <AvailableSignals P0="FRONT_INLET" P1="TEMPERATURE" P2="ACTUAL" P3="0" />
        <AvailableSignals P0="FRONT_INLET" P1="TEMPERATURE" P2="POWER" P3="0" />
        <AvailableSignals P0="FRONT_INLET" P1="TEMPERATURE" P2="DUTY_CYCLE" P3="0" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="ACTUAL" P3="1" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="SETPOINT" P3="1" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="DUTY_CYCLE" P3="1" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="ADC_ACTUAL" P3="1" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="ADC_SETPOINT" P3="1" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="SETPOINT" P3="2" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="ACTUAL" P3="2" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="DUTY_CYCLE" P3="2" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="ADC_ACTUAL" P3="2" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="ADC_SETPOINT" P3="2" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="SETPOINT" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="FLOW" P2="ACTUAL" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="ACTUAL" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="SETPOINT" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="DUTY_CYCLE" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="ADC_ACTUAL" P3="3" />
        <AvailableSignals P0="FRONT_INLET" P1="PRESSURE" P2="ADC_SETPOINT" P3="3" />
        <AvailableSignals P0="OVEN" P1="TEMPERATURE" P2="SETPOINT" P3="0" />
        <AvailableSignals P0="OVEN" P1="TEMPERATURE" P2="ACTUAL" P3="0" />
        <AvailableSignals P0="OVEN" P1="TEMPERATURE" P2="POWER" P3="0" />
        <AvailableSignals P0="OVEN" P1="TEMPERATURE" P2="DUTY_CYCLE" P3="0" />
        <AvailableSignals P0="ATMOSPHERIC_PRESSURE" P1="UNKNOWN" P2="UNKNOWN" P3="0" />
        <AvailableSignals P0="COLUMN" P1="TEMPERATURE" P2="ACTUAL" P3="1" />
        <AvailableSignals P0="COLUMN" P1="PRESSURE" P2="ACTUAL" P3="1" />
        <AvailableSignals P0="COLUMN" P1="PRESSURE" P2="SETPOINT" P3="1" />
        <AvailableSignals P0="COLUMN" P1="FLOW" P2="ACTUAL" P3="1" />
      </Fixed>
      <Variable PressureUnits="PSI" AutoPrepRunEnabled="false" ColumnOutletPressure="0">
        <Inlet Gas="H2" CryoType="UNKNOWN" FlowSensorAutoCal="0" SeptumPurgeRegulaterAutoCal="0" HeConsControlled="OFF" Sampler="ALS_SAMPLER" />
        <Inlet Gas="UNKNOWN" CryoType="UNKNOWN" FlowSensorAutoCal="0" SeptumPurgeRegulaterAutoCal="0" />
        <Detector LitOffset="0.5" FuelGas="H2" UtilityGas="Air" MakeupGas="N2" BeadVoltage="0" ElectrometerOnly="false" />
        <Detector LitOffset="0" FuelGas="UNKNOWN" UtilityGas="UNKNOWN" MakeupGas="UNKNOWN" BeadVoltage="0" ElectrometerOnly="false" />
        <Detector LitOffset="0" FuelGas="UNKNOWN" UtilityGas="UNKNOWN" MakeupGas="UNKNOWN" BeadVoltage="0" ElectrometerOnly="false" />
        <Detector LitOffset="0" FuelGas="UNKNOWN" UtilityGas="UNKNOWN" MakeupGas="UNKNOWN" BeadVoltage="0" ElectrometerOnly="false" />
        <Column Locked="true" UserDatabaseID="1" Present="true" NominalLength="30" NominalDiameter="0.00025" FilmThickness="2.5E-07" InletConnection="FRONT_INLET" OutletConnection="FRONT_DETECTOR" CalculatedLength="0" CalculatedDiameter="0" CalibrationMode="UNCALIBRATED" CalibrationDateTime="2001-03-21T22:00:00" HeaterID="OVEN" ColumnMaxTemp="325" ProgramMaxTemp="350" ColumnMinTemp="-60" Name="Agilent 19091S-433UI" Type="HP-5MS UI" SerialNumber="T330351H">
          <Tag Validity="VALID">
            <Properties Type="1" Manufacturer="1" TesterId="0" EEPROM_Rev="0" Length="30" Diameter="0.00025" FilmThickness="2.5E-07" MinTemp="-60" MaxTemp="325" ProgramMaxTemp="350" MfgDate="1556809948" PartNo="19091S-433UI" IDNo="T330351H" PhaseAbbrev="HP-5MS UI" />
            <Status RunCount="1197" TimeOverMaxTemp="90" TimeOverProgramMaxTemp="19" MaxAppliedTemp="352" OnTime="134512720" LengthCorrection="0" DiameterCorrection="0" />
          </Tag>
        </Column>
        <Column Locked="false" Present="true" NominalLength="0.17" NominalDiameter="2.5E-05" FilmThickness="0" InletConnection="NONE" OutletConnection="NONE" CalculatedLength="0.17" CalculatedDiameter="2.5E-05" CalibrationMode="UNCALIBRATED" CalibrationDateTime="2001-01-01T07:00:00" HeaterID="OVEN" ColumnMaxTemp="325" ProgramMaxTemp="350" ColumnMinTemp="-60" Name="Agilent" Type="" SerialNumber="" MaxTempOverride="350" />
        <Column Locked="false" Present="true" NominalLength="0.17" NominalDiameter="2.5E-05" FilmThickness="0" InletConnection="NONE" OutletConnection="NONE" CalculatedLength="0.17" CalculatedDiameter="2.5E-05" CalibrationMode="UNCALIBRATED" CalibrationDateTime="2001-01-01T07:00:00" HeaterID="OVEN" ColumnMaxTemp="325" ProgramMaxTemp="350" ColumnMinTemp="-60" Name="Agilent" Type="" SerialNumber="" MaxTempOverride="350" />
        <Column Locked="false" Present="true" NominalLength="0.17" NominalDiameter="2.5E-05" FilmThickness="0" InletConnection="NONE" OutletConnection="NONE" CalculatedLength="0.17" CalculatedDiameter="2.5E-05" CalibrationMode="UNCALIBRATED" CalibrationDateTime="2001-01-01T07:00:00" HeaterID="OVEN" ColumnMaxTemp="325" ProgramMaxTemp="350" ColumnMinTemp="-60" Name="Agilent" Type="" SerialNumber="" MaxTempOverride="350" />
        <Column Locked="false" Present="true" NominalLength="0.17" NominalDiameter="2.5E-05" FilmThickness="0" InletConnection="NONE" OutletConnection="NONE" CalculatedLength="0.17" CalculatedDiameter="2.5E-05" CalibrationMode="UNCALIBRATED" CalibrationDateTime="2001-01-01T07:00:00" HeaterID="OVEN" ColumnMaxTemp="325" ProgramMaxTemp="350" ColumnMinTemp="-60" Name="Agilent" Type="" SerialNumber="" MaxTempOverride="350" />
        <Column Locked="false" Present="true" NominalLength="0.17" NominalDiameter="2.5E-05" FilmThickness="0" InletConnection="NONE" OutletConnection="NONE" CalculatedLength="0.17" CalculatedDiameter="2.5E-05" CalibrationMode="UNCALIBRATED" CalibrationDateTime="2001-01-01T07:00:00" HeaterID="OVEN" ColumnMaxTemp="325" ProgramMaxTemp="350" ColumnMinTemp="-60" Name="Agilent" Type="" SerialNumber="" MaxTempOverride="350" />
        <Oven Voltage="UNKNOWN" Type="FAST" SlowFan="false" MaximumTemperature="350" />
        <PCM>
          <ChannelA Name="" Gas="UNKNOWN" />
          <ChannelB Name="" Gas="UNKNOWN" ControlMode="UNKNOWN" />
        </PCM>
        <PCM>
          <ChannelA Name="" Gas="UNKNOWN" />
          <ChannelB Name="" Gas="UNKNOWN" ControlMode="UNKNOWN" />
        </PCM>
        <PCM>
          <ChannelA Name="" Gas="UNKNOWN" />
          <ChannelB Name="" Gas="UNKNOWN" ControlMode="UNKNOWN" />
        </PCM>
        <PCM>
          <ChannelA Name="" Gas="UNKNOWN" />
          <ChannelB Name="" Gas="UNKNOWN" ControlMode="UNKNOWN" />
        </PCM>
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <Valve Name="" Type="NONE" LoopVolume="0" BCDInverted="false" StepTime="0" Inlet="UNKNOWN" />
        <AuxHeater Name="" Type="UNKNOWN" />
        <AuxHeater Name="" Type="UNKNOWN" />
        <AuxHeater Name="" Type="UNKNOWN" />
        <ALS>
          <FrontInjector SyringeVolume="10" SolventMode="0" />
        </ALS>
        <PSDEPC Name="" Gas="UNKNOWN" />
        <PSDEPC Name="" Gas="UNKNOWN" />
        <PSDEPC Name="" Gas="UNKNOWN" />
        <PSDEPC Name="" Gas="UNKNOWN" />
      </Variable>
    </GCConfiguration>
    <DSAVersion>3.9 [141]</DSAVersion>
    <SleepMethod />
    <WakeMethod />
    <ConditioningMethod />
    <Features Feature="BLANK_RUN_ANALYSIS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="COLUMN_COMP" Supported="SUPPORTED_FEATURE" />
    <Features Feature="ColumnConditioning" Supported="SUPPORTED_FEATURE" />
    <Features Feature="DET_GAS_RUNTIME_EVENTS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="EXTEND_RUN" Supported="SUPPORTED_FEATURE" />
    <Features Feature="FEATURES_SELECTION" Supported="SUPPORTED_FEATURE" />
    <Features Feature="GETHSCONFIG" Supported="SUPPORTED_FEATURE" />
    <Features Feature="GETMSDCONFIG" Supported="SUPPORTED_FEATURE" />
    <Features Feature="GUARD_CHIP_CONSUMABLE_INFO" Supported="UNSUPPORTED_FEATURE" />
    <Features Feature="LINER_INFO" Supported="SUPPORTED_FEATURE" />
    <Features Feature="MMI_POSTRUN_TOTALFLOW" Supported="SUPPORTED_FEATURE" />
    <Features Feature="OVERRIDE_OVEN_MAX_TEMP" Supported="SUPPORTED_FEATURE" />
    <Features Feature="PNEUMATIC_RTE" Supported="SUPPORTED_FEATURE" />
    <Features Feature="QUIET_MODE" Supported="UNSUPPORTED_FEATURE" />
    <Features Feature="RUNTIME_EVENTS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SAMPLE_OVERLAP" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SAMPLE_PREP" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SETHSCONFIG" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SETMSDCONFIG" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SET_IN_SEQUENCE" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SET_RECOVERABLE_ERRORS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SET_VALVE_STATUS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SIGNAL_EVENTS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SSL_DIRECT_INJECT" Supported="SUPPORTED_FEATURE" />
    <Features Feature="SYRINGE_INFO" Supported="SUPPORTED_FEATURE" />
    <Features Feature="TRAP_TRACK_OVEN" Supported="UNSUPPORTED_FEATURE" />
    <Features Feature="VIRTUAL_SIGNALS" Supported="SUPPORTED_FEATURE" />
    <Features Feature="browser_method" Supported="SUPPORTED_FEATURE" />
    <Features Feature="browser_plot" Supported="SUPPORTED_FEATURE" />
    <Features Feature="browser_sequence" Supported="SUPPORTED_FEATURE" />
    <Features Feature="data_deletion" Supported="SUPPORTED_FEATURE" />
    <Features Feature="lui_plot" Supported="SUPPORTED_FEATURE" />
    <Features Feature="lui_run_start" Supported="SUPPORTED_FEATURE" />
    <Features Feature="peak_evaluation" Supported="SUPPORTED_FEATURE" />
    <Features Feature="sequence_tab" Supported="UNSUPPORTED_FEATURE" />
    <Features Feature="time_date_change" Supported="SUPPORTED_FEATURE" />
    <Features Feature="trend_data" Supported="SUPPORTED_FEATURE" />
    <Features Feature="verification_run" Supported="SUPPORTED_FEATURE" />
    <Syringes xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/">
      <SyringeBarcodeData Position="0" StatusCode="0" />
      <SyringeBarcodeData Position="1" StatusCode="0" />
    </Syringes>
    <Liners xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/">
      <LinerBarcodeData Position="0" StatusCode="0" />
      <LinerBarcodeData Position="1" StatusCode="0" />
    </Liners>
    <GuardChips xmlns="http://www.agilent.com/schemas/Analytical/Instrument/GC7890/2006/">
      <GuardChipBarcodeData Position="0" StatusCode="-1" />
    </GuardChips>
  </configuration>
</GCContextualMethod>